Difference between revisions of "2019 December"
(Created page with "[http://www.ovsa.njit.edu/wiki/index.php/Expanded_Owens_Valley_Solar_Array#Observing_Log Back to Observing Log] == December 06 == '''12:42 UT''' We had more problems so far t...")
Revision as of 12:49, 6 December 2019
12:42 UT We had more problems so far this month involving mainly Ant 14, perhaps exacerbated by more snow. The X polarization on the high-frequency receiver showed a 15 dB loss of RF power, so Mark replaced the cable on Dec. 2. However, he accidentally left a glove at the focus, which fell down into the RX Select gear mechanism, so it was stuck at some intermediate position. The low-frequency receiver was still getting some signal, but no calibration was possible for the high-frequency receiver. Yesterday Mark realized the problem and managed to get the receiver unstuck and working again. Last night we finally got some reasonable calibration, which showed that a 5 ns delay difference was introduced. I updated the delay so hopefully starting with today's data the observations will again be nominal. Despite these problems, and the lack of a good calibration for about a week, our daily pipeline images still look pretty good!