11:03 UT The data from 19:00 UT yesterday until now are no good. Yesterday Kjell replaced the 800 MHz clock derived from the signal generator with one from a new Valon 5009 Synthesizer. Although the data seemed to be okay, I could not see any coherence in the phasecals or refcals. That prevented me from getting a delay correction. This morning I checked the delay using a communication satellite and discovered that two antennas, 8 and 14, had extremely high delays, about 300 ns! I rebooted the correlator this morning and rechecked, and now they are fine. I have seen this strange behavior once before, so I can only surmise that occasionally on a reboot of the correlator we can get really wacky delays on some digitizers (or ROACH channels--maybe not the digitizer itself). This suggests that we really need to make a quick observation of a satellite after a reboot in order to do a sanity check on the delays.